News & Updates
Search Research Content
Resource Finder at Kennedy Krieger Institute
A free resource that provides access to information and support for individuals and families living with developmental disabilities.
Printed circuit board ion trap mass analyzer: its structure and performance.
|Title||Printed circuit board ion trap mass analyzer: its structure and performance.|
|Publication Type||Journal Article|
|Year of Publication||2013|
|Authors||Jiang D, Jiang G-Y, Li X-X, Xu F-X, Wang L, Ding L, Ding C-F|
|Date Published||2013 Jun 18|
An ion trap (IT) mass analyzer can be simply built with low cost material-the printed circuit board (PCB). A printed circuit board ion trap (PCBIT) can perform ion trapping, mass analysis, and tandem mass spectrometry as a conventional ion trap mass analyzer. In a PCBIT, each PCB electrode was fabricated to specially designed patterns with several separate electric strips. The strips' electrodes were insulated from each other and applied with different voltages during the experiment. Therefore, the electric field distribution inside the ion trap region may be adjusted and optimized by simply adjusting the voltage on each strip. The performance of the PCBIT can also be optimized since the property of an ion trap is strongly dependent on the field distribution. The fabrication, operation, and performance of the PCBIT are described and characterized in this paper. A prototype PCBIT was built with two pairs of 64 mm × 12 mm PCB rectangular plates and one pair of 10 mm × 10 mm stainless steel square plates. A mass analysis with a resolving power of over 1500 and a mass range of around 3000 Th was observed. The mass-selected isolation and collision-induced dissociation (CID) of ions were also tested using the homemade PCBIT system. The adjustable electric field distribution, simple structure, and low cost of PCBIT make it certainly suitable for the further miniaturization of the portable mass spectrometer.
|Alternate Journal||Anal. Chem.|